EMAY Bluetooth Pulse Oximeter is a blood oxygen saturation (SpO2) & pulse rate monitor with bluetooth capability.
The EMAY app provided allows you to view and record the data wirelessly via bluetooth, review your SpO2 at any point in time and generate professional report to share easily.
Important Notes: * Data will be recorded in App, not in the Oximeter itself. The oximeter does not have built-in memory. * Data can be recorded only when theApp is running. * Strong movement & direct sunlight might affect the accuracy of the readings. Please stop the exercising for 10seconds and avoid direct sunlight before measuring. * The device is intended for reference only, not for medical or diagnostic use.
1.2 Launch the app and set up a profile, remember to select "yes" for all permissions when requested and enable the Bluetooth on your smart phone. (Permissions are requested by smartphone system, not by EMAY app )
1.3 Turn on the oximeter and put it on your finger to start the measuring.
1.4 Tap Connect to device on the app to start the synchronizing... Once synchronized, the readings will be displayed while data will be recorded automatically in the app.
Note: Remember to keep the app running in order to get data recorded continuously. Device and Phone must be placed within 5 meters to keep the bluetooth functioning.
Tips: The three bar graphs on each
recording gives you the general idea
of how your SpO2 & HR do over time. > Left bar shows the highest & lowest
oxygen level over time. > Middle bar shows the highest & lowest heart rate over time. > Right bar shows percentage the SpO2
is above & below the threshold.
For example, if all your SpO2 is above 94 (the default threshold)
over time, then it will be 100% above and 0% below.
1.6 Tap any recording to enter the detail interface. Review
SpO2 & PR (second by second) by sliding your finger.
Tips: As you will notice there are
two Y axis, the left Y axis is for
SpO2 % and the right Y axis is for
1.7 Tap “Report” at the right upper corner, and select
“Summary Report” to view the detailed analysis and trends
of your overnight SpO2 & HR:
1.8 Tap the icon to save or share the report by email.
1.9 Select “Data List” to enter the CSV file of granular data,
tap “Share” at the right upper corner to send the CSV file
2. How to Set the Threshold ?
The default threshold setting of SpO2 (%) is 94.
The default threshold setting of PR (bpm) is 100 at upper limit,
60 at lower limit.
Enter the setting interface to customize your own threshold
3. Print, Share or Delete Data
Press "Edit" to print, share or delete data in bulk.
4. Manage Profiles
• Edit or delete the profile by swiping it to left • Add new profile by tapping “Add User”
Unable to turn on the oximeter?
a. Make sure the two AAA batteries are
installed in the opposite directions.
b. Wiggle the battery springs and
reinstall the batteries.
c. Contact us if above doesn't help
The Pulse Oxygen Saturation(SpO2) Pulse Rate(PR) Pulse Intensity(bar-graph)
Measurement range: 30bpm~250bpm (resolution: 1bpm)
Accuracy: ±2bpm or ±2%, whichever is greater.
Internally powered equipment, type BF applied part
Range: Continuous bar-graph display, the higherdisplay indicates the stronger pulse.
2 AAA 1.5V alkaline batteries
Dimension: 58mm(L) x 32mm(W) x 34mm(H)
Weight: About 52g (including batteries)
Dimension and weight
Dimension:58mm(L) x 32mm(W) x 34mm(H) Weight:About 52g (including batteries)
5 C ~40 C
-10 C ~ 40 C
15% to 80% when working 10% to 80% storage
70kPa ~ 106kPa
Pulse wave measurement sensitivity under weak perfusion conditions:
The required test equipment (BIO-TEK INDEX PulseOximeter Tester) can measure available pulse wavesignals with an amplitude of 6% of the amplitude ofthe analog pulse wave signal.
Anti-interference ability of ambient light
use BIO-TEK INDEX Pulse Oximeter Tester to apply interference signal test, the instrument can work normally.
This product is covered by a 1-year manufacturer's warranty, and we promise to deliver great customer service for the lifetime of your device. While your warranty does not cover damage or mistreatment to your device, our priority is your happiness, and we encourage you to contact us about any issues you might have.
Your device is warranted by the manufacturer against defects in materials and workmanship for one year from the original purchaser from the date of purchase. Proof of purchase is required.
The warranty is void if the product has been subjected to mechanical damage or mistreatment, such as immersion. This warranty is in lieu of all other warranties, and limits the liability of the manufacturer. This warranty gives you certain legal rights and you may have other rights depending on which state the product was purchased in.