Portable EKG Monitor
- GETTING STARTED
- CONTACT US
- BUY THIS PRODUCT
- OUR STORY
How to Use EMAY EMG-20 Wireless EKG Monitor ?
The Emay Portable ECG is designed for family and individual users to track heart activity at home, at work or on a trip. It records heart rate and up to a 30 seconds' single channel ECG which could be used as a good reference.You can review, print and share your EKG strips/rhythm on smart phone or computer.
WARNING: This device is intended for heart activity performance reference for initial evaluation only, not for medical diagnostic use. Please contact your healthcare provider immediately or go yo nearest hospital if any cardiac symptom occurs.
2. Package Contents
ECG Monitor * 1
USB Cable * 1
User Manual * 1
3. How to Use ?
3.2 Four Ways to Record Your ECG
Choose one of the four ways to record your ECG.
W1 and W2 are the recommended ways.
Below information is very important, please read them through before
starting the recording:
IMPORTANT TIPS TO GET QUALITY AND ACCURATE ECG RECORDING:
1) Stay completely still during the recording especially your hands. We suggest you sit down and place your hands on the desk to keep them steady. Any movement might cause false diagnosis and artifacts (very important!!).
2) Ensure your skin is in full & consistent contact with the metal electrodes on both ends of the device until recording completed. Don't let go of your finger or hand during the recording.
3) DO NOT press the electrodes too hard. Gentle touch and pressure would be just fine.
4) If your skin is too dry, wipe them with disinfectant alcohol, saline or gel to strengthen the electric conductivity and recording efficiency before recording.
3.3 Start Your ECG Recording Now! (Do not use the device when it is charging)
- Turn on the device by pressing the power button firmly for 3-5 seconds.
- Hold the device using one of the fours ways mentioned to start yourpersonal ECG journey.
- Once you touch the metal electordes on both ends of the device, a fewseconds of "Orange" ECG rhythm will display, which means it's preparing.
- When the signal becomes stable, the rhythm will trun to"Green" andthe real recording begins. It will record a 30 seconds' ECG each time.
- Instant analysis & result will be displayed on the screen once therecording finished after 30 seconds (Pressing the button allows you toreview the results of past recordings)
There are 12 kins of cardiac situations that can be detected:
|No abnormalities||Indicates normal sinus rhythm with no rhythm or heart rate abnormalities detected. Your heart rates are from 60-100 beats per minute (bpm)|
|Bradycardia||A heart rate slower than 60 beats per minute. This can be normal for many people unless certain symptoms appear.|
|Tachycardia||A heart rate faster than 100 beats per minute. There are many heart rhythm disorders (arrhythmias) that can cause tachycardia. However, sometimes this can be normal response to exercise, stress, anger, fear or anxiety.|
|Arrhythmia||Indicates abnormal heart rhythm, which refers to any change from the normal sequence of electrical impulse. It means that your heart beats too quickly, too slowly, or with an irregular pattern. A-fib is one of the most common type of Arrhythmia.|
|Missed beat||Also known as skipped beat, it indicates the heart's regular rhythm is interrupted by a premature or early beat.|
|Accidental VPB||Ventricular Premature Beat (VPB), also known as Premature Ventricular Contraction (PVC). It indicates an extra heartbeat originates in the ventricle and disrupts the heart’s normal rhythm. |
Accidental (Occasional) VPB usually isn’t concern in people without heart disease. You might take attention or treatment if the VPB happens frequently.
|VPB trigeminy||Two normal (sinus) heartbeats followed by one VPB (extra beat)|
|VPB bigeminy||One normal (sinus) beat followed by one VPB (extra beat)|
|VPB couple||Two VPBs in a row|
|VPB runs of 3||Three VPBs|
|VPB runs of 4||Four VPBs|
|VPB RonT||VPB beats so early in the cardiac cycle that it falls on the apex of the preceding T wave|
|ST elevation||It refers to a finding on an electrocardiogram wherein the trace the ST segment is abnormally high above the baseline|
|ST depression||It refers to a finding on an electrocardiogram wherein the trace the ST segment is abnormally lower than the baseline|
Built-in rechargeable lithium battery, which can continuously work for more than 3 hours when fully charged. Battery recharging life up to 300 times.
Charging time: 2-4 hours by connecting with computer. 2 hours by connecting with power adapter (output current > 500mA, 5V)
- How to set the time/date of the device ?
Connect the device to the app or pc software, its time/date will be updated automatically.
- No waveform or waveform disappeared during the test?
Please contact us immediately at email@example.com
- The battery exhausts very soon?
Charge the device with USB wall charger for one hour.
- Device is hard to turn on ?
Keep pressing firmly right at the center of the power button for 3-5 seconds.
- It's difficult to pick up the heart rate and waveform? Or it takes long time to turn to "Green" official recording?
Please follow below instructions :a. Dampen your hand or contact spot with alcohol or saline before using.b. Calm and relax for 5-10 mins before using (If possible)c. Stay still as much as you can during recording.d. Do not press the metal electrodes too hard, just gentle and full touch/pressure would be fine.
- The results seem inaccurate, inconsistent or too many artifacts?
Please follow "IMPORTANT TIPS" on page 4 and measure a few timesin a row to analyze comprehensively.
Shall you have any issue or trouble with the device, please contact us immediately at firstname.lastname@example.org. Most emails will be responded within 12 hours.
1. Sync the ECG Data to Your Smart Phone
1.1 Download the App "Emay Portable ECG Monitor" from App Store or Google Play via below link:
1.2 Launch the app and enable the Bluetooth on your smart phone(Remember to select "yes" for all permissions on the App. The EMAY App is only for offline use, your information & data will be perfectly safe on your phone )
1.3 Turn on the ECG Monitor
1.4 Tap "Connect" on the app to start the synchronizing...
1.5 List of data will be displayed once the sync completed (Figure 1).
1.6 Click any data to enter the detail page (Figure 2)
1.7 Tap the icon at the right upper corner to generate EKG Report (Figure 3)
1.8 Tap the icon to save or share the report by email.
2.1 Power on the ECG device
2.2 Open the App - press"Settings"->"Device Setting" to sync with device
2.3 Here you can change the recording time or clear the memory of thedevice.
3. Manage Profile
Add new user accounts for family members (Figure 4)
Edit or delete profile by swiping it to the left (Figure 5)
4. Print, Share and Delete Data in Bulk
Press "Edit" button on the home screen to print, share or delete data in bulk.
Contact us immediately shall you have any issue or trouble with the App. We take your concern with great attention, all emails will be responded within 12 hours:
Windows : Win 7 / Win 8 / Win 10 / Win 11
Mac : OS X 10.12 or later
2. Install the PC Software
Download and install the PC Software from below link:
When you install software for Mac, please drag the"Portable ECGmonitor" icon to the "Application" folder so you can find it easily inFinder- Applications or Launchpad
3. Download & Delete the ECG Data
3.1 Connect the ECG Device with Computer
First, turn on the ECG monitor by pressing the play button for 3 seconds, then connect the ECG monitor to your computer with the supplied USBcable. Open the PC software and click "New search" the list of recordeddata will be displayed under the"Download" tab:
Important Note: You might find your PC software unable to recognize or detect the devicesometimes. Please turn on the ECG device prior to the connecting . Onlywhen the device & screen is ON can it be recognized and operated.
3.2 Download the ECG Data
You can either double-click a single data line to download, select multiple lines, then click the"Download selected" button on the right, or click the "Download all" button. All the downloaded data will be stored under"Manage" tab for review and analysis.
3.3 Device Setting
Here are what can be set for the device:
- Threshold of Tachycardia & Bradycardia ( Input the number and press"Enter" on the keyboard to reset the threshold)
- Screen display direction
- Recording time
- Heartbeat sound
3.4 Deleting Data from the ECG Device
The ECG device is able to store up to 100 records. If the storage is full, you can delete some records by clicking "Delete selected" or "Delete all" button under the "Download" tab.
4. Review & Manage ECG Data (Click "Manage" Tab)
4.1 Adding or Editing Personal Information
Select one or multiple records, right-click them and select "Modify" to enter the edit window, Here you can input your information such as "Name" "Sex" "Age" and "Diagnosis"
4.2 To Review Your EKG Rhythm
Double-click any single record to review the EKG rhythm. You can "measure" the waves, adjust "Gain, Speed and Lines" to amplify the height of rhythm, edit the diagnosis field, enter doctor's name and print the report.
4.3 Export Summary List
Click the "Export" button in the middle right under "Manage" tab to export listing of all records in ".csv" so you can check the summary of all records with information of Name / Sex / Age / Time length / Check Time / Heart Rate / Feature / Diagnosis / Doctor
5. Share Your ECG Report
5.1 Print It Out
If there's printer connected with your computer at home, just double-click the tracing you want, click "Print" in the bottom right.
5.2 Save the report as XPS/PSD and email it.
For Windows: If you do not have a printer at home, the report can be saved as XPS/PDF copy to email your doctor or print elsewhere. First, double-click the selected tracing, click "Print" in the bottom right, here you can choose "Microsoft XPS Document Writer" (Win7, Win10) or "Microsoft Print to PDF" (Win10), then select "Print" to create XPS or PDF document.
For MacOS: If you do not have a printer at home, the report can be saved as PDF to email your doctor or print elsewhere. First, double-click the selected report, click "Print" in the bottom right, then click "PDF", here you can select "Save as PDF"
- Software Installation Fails
If your computer can not read the USB flash drive, please download it from our website directly. And if you need help install the software, please contact us at:
- Software can't recognize/detect the ECG device ?
Please turn on the device by pressing the play button for 3 seconds, and click "New search" quickly within 10 seconds. Only when the device & screen is ON can it be recognized and operated. If above instruction doesn't work, plug and unplug the cable for 2 times OR use another usb cable to try again. If neither of above helps, please contact us immediately at email@example.com, we will help you fix it.
- Waveform/Tracing Disappeared
If you have started the software, downloaded the recordings, and found that there's only heart rate data without any waveform, please contact our service team immediately and we will fix it for you.
- Download failure or only part of the reports was downloaded?
There are multiple reasons for this, please try below measures:
- use another USB cable to connect the device to the PC software to avoid a USB cable defect. Any standard micro USB for smart phone should be fine.
- try another USB port on the computer in case the one you often used is loose or defective.
- Uninstall and reinstall the software to make sure the software is well installed
- it could also be hardware defect, if all above methods do not work, please contact us for replacement or return.
- The "Check time" shows "2015-01-01"?
The "Check time" of the first couple recordings might show "2015-01-01", but once the device is initially connected to your computer, the time & date of your future recordings will be correct.
- Can I download the same report to both phone app and PC software ?
If you download the report to the app on phone first, you can still download it again on pc software, just remember to tick "All" button on the top of the interface on pc software. However, if you download the report to pc software first, then you won't be able to download again on phone app. That's why we suggest downloading all reports to your smart phone first.
|Lead||Standard 3 leads, I I I||Calibration voltage|
|Standard sensitivity||10mm/mV+5%||Amplitude frequency|
|Input impedance||250MQCMRR: ≥60dB||Scanning speed|
|Type of protection against electric shock||Internal power device||Metal electrode degree of protection against electric shock|
|Degree of waterproof||IP22||Display|
|Size||100mm (L)*45mm (W)*15mm (H)||Weight|
|Service Life||3 years|
This product is covered by a 1-year manufacturer's warranty, and we promise to deliver great customer service for the lifetime of your device. While your warranty does not cover damage or mistreatment to your scale, our priority is your happiness, and we encourage you to contact us about any issues you might have.
Your device is warranted by the manufacturer against defects in materials and workmanship for one year from the original purchaser from the date of purchase. Proof of purchase is required.
The warranty is void if the product has been subjected to mechanical damage or mistreatment, such as immersion. This warranty is in lieu of all other warranties, and limits the liability of the manufacturer. This warranty gives you certain legal rights and you may have other rights depending on which state the product was purchased in.
If your scale is defective, please contact our service team: